This paper demonstrates the way of how the value of any quantity used in Microwaves and RF and defined, either directly or indirectly, by one ore more complex differentiable functions of one or more Vector Network Analyzer VNA measurements, is affected by its ubiquitous Differential Errors DEs, when these DEs are expressed in terms of the manufacturers' data for both the unavoidable uncertainties of the calibration standards and the inescapable inaccuracies of the VNA indications. This is illustrated by a step-by-step graphical construction of the complex DE Regions DERs and their real rectangular and polar DE Intervals DEIs, which the authors introduced and used in all previous parts of this series. As it is geometrically shown, in order to build the composite figure of any complex DER and then of its DEIs only three basic types of simple figures are needed. These figures result by using combinations of appropriate complex transformations, which are well-known from Complex Analysis, that is: translation, scaling (stretching or shrinking), and rotation. A number of selected applications are presented and some useful, general conclusions are drawn.
Comments: 19 Pages. FunkTechnikPlus # Journal, Issue 16 - Thursday 31 May 2018 - Year 5, CC-BY 4.0, pp v1-7--v1-25
[v1] 2018-05-26 04:54:06
Unique-IP document downloads: 12 times
Vixra.org is a pre-print repository rather than a journal. Articles hosted may not yet have been verified by peer-review and should be treated as preliminary. In particular, anything that appears to include financial or legal advice or proposed medical treatments should be treated with due caution. Vixra.org will not be responsible for any consequences of actions that result from any form of use of any documents on this website.
Add your own feedback and questions here:
You are equally welcome to be positive or negative about any paper but please be polite. If you are being critical you must mention at least one specific error, otherwise your comment will be deleted as unhelpful.